beauty equipment factoryDEC.18,2021
Key words: ultrasonic facial machine manufacturerDEC.15,2021
Is Slimming Machine Safe To Use?DEC.03,2021
2 IN 1 BEAUTY INSTRUMENTDEC.03,2021
What are the benefits of Electric Massage Beds?DEC.01,2021
2 IN 1 BEAUTY INSTRUMENTDEC.01,2021
Super oxygen injection apparatusNOV.26,2021
The product of the B-8635 descriptionNOV.19,2021
Matters needing attention
VERSION: 20090808 USER MANUAL
Overview for function mechanism of the Ultrasonic Beauty Instrument……………1
Treatment and beautifying function of the Ultrasonic Beauty Instrument ……… 1
Treatment dose and waveform selection of the Ultrasonic Beauty Instrument …… 3
Graphical representation of the operation panel ………… ……………………… 3
Steps of the ultrasonic beautifying method………………………………………… 4
Matters needing attention ………………………………………………………… …4
Technical indexes of the comprehensive beauty instrument…………………………5
Notice for use…………………………………………………………………………6
Technical indexes of the comprehensive beauty instrument
The ultrasonic frequency of this instrument is ±10% of 1MHz; scope:1-30 minutes (one-minute stepping, adjustable); the number of the display stages for the single-step adjustment power output is 10，specifically, three stages for green, three stages for yellow, and four stages for red;
Output power: 0-15W, adjustable;
Pulse output modulation frequency: 4Hz;
Output voltage: 110V/220V (adjusted by the power transfer switch behind the instrument body), 50-60Hz
Power of supply＜40W;
Notice for use
We appreciate your order this product! Please go though this manual before use in order to ensure the maximum performance of the instrument and the safety in use. Please keep properly for further requires
Ø1.It is not allowed to be used for the pregnant women;
Ø2.It is not allowed to be used for the cardiac patients;
Ø3.It isnot allowed to be used for the hypertensive patients;
Ø4.It is not allowed to be used for the patients Sufferinginternal injuries;
Ø5.It is not allowed to beused for the patients suffering skin diseases;
Ø6.It is not allowed to be used for the people withopenwounds on the treatment parts.
1.Power button 2.Waveform mode selectiom button 3.Handle mode selection mode
4.Increase of power 5.Decrease of power 6.Increase of time 7.Decrease of time
Steps of the ultrasonic beautifying method
1.Beforethe application of the ultrasonic instrument, it is necessary to clean thefacial skin with the appropriate facial cleanser;
2.Afterthe facial cleanser has been wiped out, it is necessary to steam the face withthe spray device for 10 minutes;
3.Clearout the facial blackhead, acne, etc;
4.Gentlypat the appropriate shrinkage water on each part of the face;
5.Gentlywipe out the shrinkage water;
6.Clearlyanalyze the skin nature and the problem to be dealt with, and then select the most suitable ointment.
7.Applythe selected ointment on the face properly and evenly;
8.Switchon the power and select an appropriate output socket to connect with the ultrasonic probe;
9.Adjustthe power size (ENERGY) according to the power switch (START);
10.Adjust the timeappropriately, which generally will be 15 minutes;
11.Select the handle (SELECT)mode, that is, round head () or sharp head () according to the need; waveform(STATE)-“continuous wave () or impulse wave ()”;
12.Massage the skin applied with ointment with the probe;
13.After the ultrasonic wavetreatment has been finished, it is necessary to clean the probe;
14.If the appropriateChinesetraditional medicine powder is selected, the skin will become fresh andpleasant.
Matters needing attention:
1.Before the ultrasonic wave treatment, it is necessary to clean the face (or the affected part);
2.The applied medicine shall have a certain viscosity so that the ultrasonic working head can be perfectly combined with the skin, which is conducive to the skin absorption; meanwhile, to prevent the ultrasonic working head from being burnt out, it is necessary apply enough medicine on the skin.
Overview for function mechanism of
the Ultrasonic Beauty Instrument
Since the introduction of the ultrasonic beauty instrument technology in the 80s, due to the wide treatment range, quick effect, high curative rate and easy operation, it has got a wide attention, popularization and promotion, which has made it become the most practical and popular equipment among so many beautifying devices. Specifically, its functions mainly lie in three aspects:
The mechanical function of the ultrasonic wave lies in the functions to soften the tissue, enhance the penetration, improve the metabolism, facilitate the blood circulation and stimulate the nervous system & cell.
The warming function of the ultrasonic wave can cause the changes of the blood vessel function and metabolic process, can strengthen the blood circulation to lower down the exciting point of the nerve, can produce the analgesic effect, as well as can relax the spastic skin and play the spasmolytic
The ultrasonic wave is featured by the functions to strengthen the catalysis and accelerate the metabolism, which can change the PH value towards the alkalinity and facilitate the medicine to permeate into the bacteria so as to improve the medicine’s sterilization ability.
Treatment and beautifying function of the Ultrasonic Beauty Instrument
1.The ultrasonic medicine penetration treatment function is characterized with the following features:
(1) The medicine is not limited to ionization material and water soluble substance so that the medicine source is broad;
(2) Compared with the pure ointment application, the ultrasonic medicine penetration can be absorbed more and deeper;
(3) The medicine concentration is not limited by the dissociation degree, etc;
(4) There is no such problem as impact strength and polarization of practice;
(5) There is no such phenomenon as electrical stimulation so that the accident of electric injury will not happen;
(6) It is easy and simple to handle; meanwhile, the applied medicine shall have a certain viscosity.
(1) Softening the blood clot and eliminate the “red face”:
To combine the ultrasonic pulse treatment with the appropriate ointment for promoting blood circulation to remove blood stasis can rectify the deformed imperceptible blood vessels and make them return to normal thus to eliminate the red face.
(2) Eliminate the scar left by the healed acne
Because the ultrasonic wave can penetrate the skin, which not only can dissolve the dry blood oozed into the skin, but also can strengthen the blood circulation and metabolism, it can make the color print fade away faster; generally, the treatment combined with the ultrasonic wave and the acne cream shall be carried out once every other day and ten times will be one course of treatment.
(3) Eliminate the skin pigmentation disorder
To combine the ultrasonic wave with the Chinese traditional medicine cream or freckle cream and the high-unit vitamin C will produce effect quickly and thoroughly eliminate the skin pigmentation disorder; specifically, the treatment with ultrasonic wave applied will produce the desired result in
three times and 10-40 times of treatment will eliminate it.
(4) Dissolve the pigment to eliminate the subcutaneous plaque
The practice will be calculated by the course of treatment and ten times will be one course of treatment, among which one shall take a rest for seven days; the uniqueness shall be credited to the fact that the ultrasonic impact can destroy the inner membrane of the pigmentation cell to prevent its formation and dissolve the pigment so as to make the pigment disappear completely with the metabolism.
(5) Remove wrinkle, remove blood stasis and conduct face lift
To apply the mechanical massage function of the ultrasonic wave, for one thing, it can adjust the change of the subcutaneous cell to go through the cell membrane; for the other thing, it can strengthen the blood circulation and metabolism; therefore, it can make the wrinkle fade away gradually and even can thoroughly eliminate the blood lump generated for being bruised by the hard object.
(6) Remove eye bags and dark circles
Due to the mechanical massage function of the ultrasonic beauty instrument, it can accelerate the blood and lymph circulation; furthermore, with the thermal effect, it can dissolve fat and accelerate skin absorption to dissipate the excessively accumulated water and fat and thus to remove the eye bags and dark circles. Use the small head of the ultrasonic beauty instrument to carry out the massage along the skin texture: five minutes for each eye bag once; the applied force shall be gentle; and the ultrasonic head shall not be directed at the direction of eye.
(7) Soften hard mass for defatting and acne treatment
Combine the ultrasonic beauty instrument with the acne pathogenic fire reduction ointment to gently massage the acne gently about one minute, which can heal the acne and avoid the formation of the resulting scar at the same time.
(8) skin sclerosis and snake skin pathosis
Combine the ultrasonic beauty instrument with the glycerol and take the liquid paraffin or high-grade emollient to carry out the massage therapy can quickly improve the symptoms.
(9) Conduct massage frequently to improve skin quality
To use the ultrasonic beauty instrument to carry out properly processing can eliminate patch and acne quickly, as well as can improve or restore the skin quality.
Treatment dose and waveform selection of the Ultrasonic Beauty Instrument
Graphical representation of the operation panel