beauty equipment factoryDEC.18,2021
Key words: ultrasonic facial machine manufacturerDEC.15,2021
Is Slimming Machine Safe To Use?DEC.03,2021
2 IN 1 BEAUTY INSTRUMENTDEC.03,2021
What are the benefits of Electric Massage Beds?DEC.01,2021
2 IN 1 BEAUTY INSTRUMENTDEC.01,2021
Super oxygen injection apparatusNOV.26,2021
The product of the B-8635 descriptionNOV.19,2021
Ⅰ. Product overview
OPT perfect pulse technology applies three-dimensional technology concept, i.e. energy + pulse width + pulse waveform, making the output of pulse sent out stable and uniform so as to make the hair removal fast, safe and painless, with functions of freckle removing and whitening. In particular, the use of OPT + E light + laser + ice wave face lift, namely the integration of four kinds of beauty functions, can effectively eliminate all kinds of skin pigments, and is capable of washing eyebrows, removing freckles and moles and Erb-YAG Laser skin beauty technology. Therefore, the machine has multiple and comprehensive functions.
1.1 Pox and acne removing
Use short-band hard light to kill propionibacterium within the acne, and utilize combination functions of red light, infrared light and radio frequency current to eliminate pigmentation while reducing sebaceous gland activity, stimulating deep collagen hyperplasia as a way to achieve a good acne removal effect.
As the composite energy of IPL and RF acts on different depth of skin without relying on a single energy, the safety is guaranteed.
1.2. Whitening skin, couperose and freckle removing
Equipment principles for pure IPL technology and the combination technology of IPL and RF are different. When the IPL technology is used only, the principle is to utilize the selective absorption pyrolysis of hemoglobin and pigment group to light, making the temperature of target tissue rise and appear necrosis to achieve the purpose of beauty without damaging normal skin tissue, especially epidermal tissue.
When the combination technology of IPL and RF is applied, the principle is to utilize the selective absorption of skin tissue to hard light to cause impedance differences of target tissue and normal skin tissue. In addition, strengthen the absorption of radio frequency current by target tissue and rely on the radio frequency to reach operation effects.
The applications of pure IPL technology and the combination technology of IPL and RF are safe in terms of eliminating epidermal plaques and small couperose skin, and whitening and skin rejuvenation. The pure IPL technology has few effects in the premise of ensuring safety, while IPL and RF combination technology has more obvious effects while guaranteeing safety.
1.3. Wrinkle removing and face lift
Adopt IPL and RF combination technology, that is, E-ray using electromagnetic waves and IPL thermal effects to provide a ideal temperature rise to stimulate the corium layer, have effect on the entire corium layer and connective tissue and stimulate collagen fibers at different depths, thus achieving the effect of removing wrinkles, face lift and non-invasive face lift.
As the 15MM radio frequency current penetrating subcutaneous acts on the corium layer and pose small effects on the epidermis with its main energy applied to the target tissue, the safety is high.
1.4 Permanent hair removal
E-light hair removal is a new long-term hair removal technology. Its principle is that filtered strong pulsed light is easily absorbed by the melanin in hair and hair follicles and converts into thermal energy which makes the temperature of hair shaft and hair follicles rise quickly. Besides, when the temperature reaches more than 60 ℃, the hair would fall off, hair follicles would suffer from denaturation, hair papilla would be damaged and gonotokont of hair would die, finally making the original hair fall off and new hair grows slowly, become sparse and light until stop growing.
Utilize the energy of 15 mm radio frequency based on the application of the IPL technology, directly acting on melanin cells and making the cells absorb more energy, thus having a great effect on removing small, light, even gold and white hair.
Ⅱ. Technical parameters
2.1. OPT+E light technical parameters
|Operation screen||8.0 inch true color touch screen|
|First pulse width||1-20ms|
|The number of pulses||1-6|
|Cooling system||Semiconductor refrigeration, closed cycle forced water cooling, air-cooled heat exchange|
2.2. Laser technical parameters
|Laser type||Gem adjustment QND：YAG laser|
|Monopulse energy||≤200mj 1064nm|
|Q-switched mode||Passive Q switching|
|Facula diameter||1-5mm continuously adjustable|
|Cooling method||Water cooling + air cooling|
Ⅲ. Instrument advantages
Ⅳ. Instrument characteristics
Ⅴ. The operation range of E light
Ⅵ. Standard configuration of instruments
Ⅶ. Installation instructions
7.1 Installation steps of the filter
7.2. Instrument installation instructions
7.2.1 Installation procedures
As the instrument is equipped with super-cooling system, the instrument shall run for 5 minutes after the first time of water adding. After that, water shall be added to the instrument again under the halt condition, until there is water flows out of the spillway hole. When the instrument is used frequently, it is recommended to change water on the 15th day. If the gun head is hot (there is air in the gun head, causing water is not circulated) within 10 seconds before the starting up a new instrument, please shut down the instrument immediately, lift the gun head over your head to shake three times and start up after water flows into water tanks. If the gun head is still hot, steps stated above can be repeated and remember not to start up without water.
Ⅷ. Operation instruments
8.1 Startup picture
8.2. Operation mode selection interface
8.3. The picture after entering OPT-E interface
8.4、OPT—E operation interface
|1||IPL energy||Regulate the output energy of the light head, and the current value of would be displayed on the energy display window during the regulation process. The value regulation range is 1~50J/cm2 and the value can be directly regulated during operation, pressing the “△ or ▽” button. Pressing the “△” button is to add energy and pressing the “▽” is to reduce energy. The output energy can be regulated quickly by pressing the “△” or “▽” button;|
|2||RF energy||Regulate the output energy of RF, and the current value of would be displayed on the energy display window during the regulation process. The value regulation range is 1~30 and the value can be directly regulated during operation, pressing the “△ or ▽” button. Pressing the “△” button is to add energy and pressing the “▽” is to reduce energy. The output energy can be regulated quickly by pressing the “△” or “▽” button;|
|3||Refrigeration regulation||It is used to regulate the refrigeration temperature of the light head with 5 gears adjustable. The more the blue snowflake, the stronger the refrigeration effect. When the blue snowflake disappears, the refrigeration is shut down;|
|4||Standby||Standby button, press the button to enter into the standby state|
|5||Ready||Ready button, press the button to make the motor to enter into the ready state|
|6||Operation records||Display (3 E light operation modes) the current operation record and you can record different operation parameters to apply to different patients. operation records can be changed by clicking 3 options;|
|7||Return buttons||It is used to return to the operation mode selection menu. When it is gray, it is not available (operation status);|
|8||Return button||operation parameters recorded in operation mode;|
|9||Light head state||Display the operation mode currently selected;|
8.5. E-Light mode parameter modification
In the operation interface, we can adjust the operation parameters. Click the parameter required to be modified on the “operation parameter setting”, then corresponding parameters would be selected and the up and down arrows would appear. When arrow pointing up is pressed, the value would increase. When the arrow pointing down is pressed, the value would decrease. The values can be saved by clicking operation parameters after the adjustment.
The operation parameters represent the data that can be adjusted during operation. By adjusting the operation parameters, we can change the energy, pulse width and pulse number of the output of the light head.
There are 8 adjustable operation parameters in total (see Figure 4). Adjust the pulse interval of the operation pulse with a adjustment range of (1 to 4s) seconds and the interval can be adjusted by clicking directly. Adjust the number of operation pulse with the adjustment range of (1 to 6) and the number can be adjusted by clicking directly. Adjust the action time of RF with the adjustment range of (50 to 2000) ms and the unit of 50 ms. The time can be adjusted by clicking directly. The greater the number, the longer the action time of RF. It is recommended to adopt the time of 300 - 900 ms and the time may be adjusted according to actual conditions.
The first pulse width: Adjust the pulse width of the first pulse. It shall be the pulse width when the single pulse is applied and it shall be the width of the first pulse when multiple pulses are applied. The adjustment range is (1 to 10 ms) with 0.1 ms of step.
The first pulse interval: Adjust the delay between the first pulse and auxiliary pulse (the second pulse) with the adjustment range of (1 ~ 100ms) and 1ms of step.
The second pulse width: Adjust the pulse width of the auxiliary pulse (the second pulse) and it is not displayed when the single pulse is applied. The adjustment range is (1 to 10 ms) with 0.1 ms of step.
The second pulse interval: Adjust the delay between the auxiliary pulse (the second pulse) and the repetitive pulse (the third pulse and the subsequent pulse). It is not displayed when the single pulse is applied. The adjustment range is (1 to 100ms) with 1 ms of step.
Repetitive pulse width: Adjust the pulse width of the repetitive pulse (the third pulse and the subsequent pulse). It is not displayed when the single pulse is applied. The adjustment range is (1 to 10 ms) with 0.1 ms of step.
Pulse output diagram
Press the "Ready" button (green) to trigger the light head and charge the capacitor automatically. When the charge is complete, the instrument enters into the operation state. Press the "Standby" button (red) to turn off the light head and enter into the standby mode. After the charging of the power supply fishes, please wait about 10 seconds and you can start to operate after the temperature of the light head drops. Press and hold the “work button” of the light head, and the light head begins to work according to operation parameters set. When the “work button” is pressed and held, the system does not reflect any other buttons and operations can be carried out only after the “work button” is released. During operation, the operation parameters can be modified. After the finish of the operation, press the “Standby” button to exit the operation procedures and turn off the light head.
When RF is operated, the good contact between RF electrodes and skin must be guaranteed with epoxy resin coated. The energy of RF shall be adjusted from low to high and the contact time of the electrodes with skin should not be too long as so to avoid scalding the skin. The RF operation is mainly felt by feeling hot inside the skin. If a strong sense of acupuncture is felt, it is too little epoxy resin coated or the counter electrode is badly contacted with skin. Under such circumstances, the epoxy resin shall be coated again or the position of the counter electrode shall be adjusted.
RF action time can be adjusted according to needs, but shall not be too long (between 300-900ms recommended). If the action time is too short, the effect is not obvious. If the time is too long, it is easy to hurt the skin.
8.7. The picture after entering the LASER interface
8.8. LASER operation interface
|1||Energy||Regulate the output energy of the light head, and the current value of would be displayed on the energy display window during the regulation process. The value can be directly regulated during operation, pressing the “△ or ▽” button. Pressing the “△” button is to add energy and pressing the “▽” is to reduce energy. The output energy can be regulated quickly by pressing the “△” or “▽” button;|
|2||Standby||Standby button, press the button to enter into the standby state|
|3||Ready||Ready button, press the button to make the motor to enter into the ready state|
|4||Return buttons||It is used to return to the operation mode selection menu. When it is gray, it is not available (operation status);|
|5||Frequency||It is used to adjust the frequency|
|6||Operation records||Display the current operation record and you can record different operation parameters to apply to different patients.|
|7||Light head state||Display the operation mode currently selected;|
Ⅸ. Operating instructions
1) Starting up: Open the instrument power switch, then the instrument display is on and the system circuit immediately enters into the self-test state. The operator, in this case, shall let the instrument run for 5 minutes before proceeding to the next operation.
Note: If the power switch is turned on and the display is still off, please check if the emergency stop switch is reset. Rotate the emergency stop switch in the correct direction to reset the emergency stop switch, and the instrument enters standby mode.
2) Preheating: preheat the instrument for at least three minutes and do not turn on the lights.
Purpose: To fully circulate the water in the water pipe, so as to avoid the light head lance burning caused by multiple times of blocked circulation, and ensure the safety of operation.
Purpose: Hair is black and would absorb a part of light during operations. Therefore, the skin would absorb relatively little light, which would reduce the care effects. The absorption of light by hair would increase the sense of stabbing pain of the people receiving the care. So it is necessary to shave hair, otherwise the small villus, after absorbing light would keep a part of heat on the skin epidermis, which may increase the skin's sensitivity.
5) Hypersensitive tests: Adjust the safety parameters and do two to three facula on the auditory faculty and observe three minutes.
Function: Do not operate if any allergic phenomenon occurs
Ⅹ. Matters need attention when operating
Observe customers’ skin condition and spot color depth, and adjust the parameters according to the customers’ bearing capacity and skin metabolism conditions. The energy shall be slightly lower than the parameters provided on the instructions when operating on customers for the first time, making customers adapt to the energy of the instrument. It is recommend to adjust the light energy from 20 for the first time with 2 to 3 increases and decreases. And the electricity energy recommended to be adjusted from 15, which can be operated several times. For the second time, the operator can communicate with customers in advance to make the customers prepare to receive the care treatment psychologically, and then adjust the energy to a higher level to operate. In addition, the operator shall tell customers to do postoperative care.
Ⅺ. Care after operations
Ⅻ. Fault phenomena and troubleshooting methods
|Fault phenomena||Cause analysis||Troubleshooting methods|
|The machine is not energized||Power supply is not input and is not supplied into the instrument, which is possibly caused by
1) Blown fuse 2) The emergency is not released
3) The electric key is not open
|1) Replace fuse
2) Release emergency stop
3) Replace the electric key
|The light head does not refrigerate after staring up||1) The refrigeration power supply is damaged
2) The instrument core plug is badly connected
3) Semiconductor refrigeration is damaged
4) There is air in the gun head, causing no circulation of water routes
|Contact manufacturers for maintenance|
|The light head leaks||1) The screw inside the light head is loose
2) Light head internal pipe falls off
3) Light or crystal breaks
|Remove the light head and contact manufacturers for maintenance|
|Lights do not light up||1) The light head is damaged
2) The front panel (controller) is damaged
3) The power supply is damaged
|Consult manufacturer by telephone to judge whether parts are damaged and required to send to repair|
|No flash (no discharge)||1) The manual switch is badly contacted
2) Operation errors
3) The controller is broken
4) The power is broken
5) The number of use of gun head is run out or its service life expires, and it shall be replaced with a new gun head
|Consult manufacturers by telephone to judge whether parts are damaged and required to send to repair|
XIII. E light technical parameters adjustment table
|Item||Skin Characteristics||IPL energy||RF energy||First Pulse Width||Pulse Number||The second pulse width||Pulse delay||Frequency|
|480nm acne||Light acne||20-30||7-11||8-10||3-5||9-12||12-15||2|
couperose skin removal
|The thicker the blood vessels, the longer the first pulse||20-30||7-11||8-10||3-5||9-12||10-12||2|
|The thicker the blood vessels, the shorter the first pulse||20-30||7-11||7-11||3-5||9-12||10-12||2|
|The flusher the face, the longer the first pulse||20-30||7-11||8-10||3-5||9-12||10-12||2|
|The darker the skin, the smaller the energy and the longer the interval||20-30||7-11||8-10||3-5||9-12||10-12||2|
|The whiter the skin, the greater the energy and the shorter the interval||25-35||7-11||8-10||3-5||9-12||10-12||2|
|The rougher the skin texture, the longer the first pulse||20-30||7-11||8-10||4-6||9-12||10-12||2|
|The finer the skin texture, the longer the first pulse||25-35||7-11||8-10||4-6||9-12||10-12||2|
|The thicker and the better the skin, the greater the energy||25-35||7-11||8-10||3-5||9-12||10-12||2|
|The thinner and the worse skin, the smaller the energy||20-30||7-11||8-10||3-5||9-12||10-12||2|
|Eye bags, wrinkles, large pores||15-25||7-11||8-10||4-6||9-12||10-12||2|
|The darker the skin, the smaller the energy and the longer the interval||20-30||7-11||8-10||4-6||9-12||10-12||2|
|Light speckle color with bad skin||25-35||7-11||8-10||4-6||9-12||10-12||2|
|Dark speckle color with good skin||20-30||7-11||8-10||3-5||9-12||10-12||2|
|Dark speckle color with bad skin||20-30||7-11||8-10||3-5||9-12||10-12||2|
|Light speckle color with good skin||25-35||7-11||8-10||4-6||9-12||10-12||2|
|The whiter the skin, the greater the energy and the shorter the interval||25-35||7-11||8-10||4-6||9-12||10-12||2|
|The darker and coarse the hair, the shorter the first pulse and the longer the interval||25-35||7-11||8-10||3-5||9-12||10-12||2|
|The lighter and finer the hair, the longer the first pulse and the shorter the interval||25-40||7-11||8-10||4-6||9-12||10-12||2|
Note: The above parameters are for reference only. The energy of operation for the first time shall be slightly smaller than the above energies and facula tests shall be carried out before the first operation. The light energy is recommended to be adjusted from 10 with increases and decreases according to customers’ actual conditions. The increase and decrease range shall be 2 to 3. If the energy is felt to be not enough in later operations, the energy increases can be considered by referring to the parameter table. The operation on each part shall not be repeated no more than two times and the operation can be conducted one more time when the energy is small. Remember that the energy can not exceed 10.
Product maintenance and after-sales service
The failures caused by unauthorized disassembly and modification;
The failures caused by inadvertently beat and fall;
The failures caused by lacking reasonable maintenance;
The failures caused by not following correct guidance of the instructions.
Instrument repair registration table
（This is attached to the instrument for repair）
Thank you for purchasing our company’s equipment. If there is maintenance need, please fill out the maintenance form. The information you provide is very important to us! Our company will repair the instrument as soon as possible after receiving.
|Instrument name：||Purchase date:|
|Name of customer:||TEL:|
|Address where the instrument returns after repairing:|
|Failure description of the instrument:
(MM) (DD) (YY)
|Name of accessories||Unit||Quantity|