- Absorb Black Acne
- High-Frequency electrotherapy
- Facial Brush
- Remove Spot
- work/stop switch vacuum & spray section.
- indicator of vacuum & spray
- intensity adjuster of vacuum & spray
- work/stop switch in H.F section
- working indicator of H.F.
- intensity adjuster of H.F.
- intensity display of H.F.
- work/stop switch in brush section
- working indicator of brush.
- intensity adjuster of brush
- intensity display of brush
- work/stop switch in remove spot section
- working indicator of remove spot
- intensity adjuster of remove spot
- intensity display of remove spot
- work/stop switch in ultrasonic section
- working indicator of ultrasonic
- intensity adjuster of ultrasonic
- intensity display of ultrasonic
- right/left select
- display of right/left
- change of wave type
- display of wave type
- the timer of ultrasonic
- time display of ultrasonic
- Vacuum & Spray Section:
- To thoroughly get rid of all dirt, impurities fat from the deep pores of the skin.
- To make the suction/spray medicine water to spray as mist to nourish and calm skin.
- Turn on the power on the section of Vacuum & Spray.
- Adjust the vacuum intensity by rotating the intensity valve.
- Turn off the power after completion.
- There are three types of glasses of which should be used separately according to any specific parts of the body.
- After the operation is over, the used glass tubes should be cleaned and sterilized and kept dry.
II、High-Frequency Electrotherapy Section:
- Use special Ultraviolet rays to treat the skin that sterilize the skin and accelerate wound healing.
- To get rid of spots, pigmentation, pimples and acne.
- Connect the plug with the corresponding socket.
- Plug the ultraviolet tube into the handle.
- Turn on the power in H.F. section and turn the button of energy lever to choose appropriate level then begin to operate.
- After the operation is over, turn off the power, pull out the ultraviolet tube and clean well and keep dry.
- Take off all metal decorations (earrings, necklace etc.) before operation.
- During the operation, not to stay too long time in a treated section to avoid scars.
- The electrodes should be kept clean and dry anytime.
III。Brush Skin Section:
- Function: Nourish skin, clean face, brush skin, massage and can improve the quality of the skin with wrinkle and acne..
- Insert the plug into the corresponding socket.
- Insert the brush into the handle.
- Turn on the power in the BRUSH section and select the proper intensity and begin to operate.
- when over, turn off the power and pull out the brush.
- Preparation before operation:
1). Connect the handle with the machine.
2). If the beautician wear the stelization glove, should clean hand and sterilize with alcohol.
3). Fix the sterilized spot pin on the handle.
4). Clean the treated section with alcohol.
5). Rub the treated section with anaesthetic for 3 minutes, and then begin operation.
- Turn on the power of remove spot section and adjust to the weakest intensity.
- Test if there is current with metal.
- Determine the current capacity according to the treat section and adjust it to meet the customer’s requirement.
- When operating, care the currency lever and adjust a proper energy lever to avoid hurt.
- Spot:special care to test and analysis spot’s conditions, some spots on the surface of skin but some spots in the deep under skin issue. If spot small, might remove it one time, if spot big and pigments deposits deep, should remove it 2-3 times. Don’t cut through and break the spot’s surface one time, should burn and remove with small frame way. To spot in the deep under skin, should spread proper cell cream on it before the wounded cure well 2~3 days, to avoid inflammation or skin hollow happen. When spread the cream, if the wounded inflame, don’t spread it.
- Caruncle:gently touch the end with pin and kill the caruncle cell, later about 6-7 days, the caruncle drops. If the caruncle is too big, should remove it several times and not finish it in one time to avoid the wounded covers bad.
- Tattoo: ordinary tattoo section is big, should choose #3 big bin and treat several times, remind not finish treatment. Otherwise, the wounded section is very easy to be inflamed, if happen, should take the sick to hospital to treat soon.
- The pin should be sterilized before use.
- Beautician should keep the handle balance not shake to avoid burning the wounded
- Soften the thrombus and remove “red face”.
- Remove scar over.
- Eliminate abnormal skin pigment.
- Lift skin to remove wrinkle.
- Remove eye pouch and black eye.
- Meliorate skin conditions.
- Before operation, first clean face well with clean facial cream.
- Wash off the cream then steam face for 10 minutes.
- Clear over the comedo on the face.
- Spread shrink water on the face and tap it gently then wipe off it.
- Test skin conditions then spread the suitable medicine cream on the face.
- Turn on the machine then adjust the power of probe to fit skin well.
- Set timer to show 15 minutes the normal work time.
- Choose wave type (sequentialwave or pulse wave) as per requirement.
- Massage on skin covered medicine cream with probe.
- After over the operation, turn off the machine and clean the dirty probe.
- If the herb powder can be available and applied on the treated section , it will have better effect.
- First clean face (or wounded section) before operation.
- The medicine cream used should be mucous to be absorbed in better by skins, at the same time to avoid burning the probe.
- Work time set to 15 minutes, no better result is obtained if prolong the time.
- if the treat section is small, use the small probe and the time is set to 8-10 minutes.
- Don’t make the probe close to the eyes or touch eyes to avoid hurt.
- Don’t make the probe in work but not operate to touch skin, as it is easy to burn the probe for over heat.
*Special attention: don’t pull the socket or the wire of the probe end to avoid damaging the internal wire when using the ultrasonic probe.